After you have finished a measurement series, rinse the measuring cell with warm distilled or deionized water. To this end, fill the supplied syringe with warm distilled or deionized water, and rinse it through the measuring cell and the waste hose. After rinsing the measuring cell with distilled or deionized water, it is not necessary to dry the cell because the water will evaporate without residues.
In addition, clean the measuring cell once a week with a suitable laboratory cleaner (e.g. Mucasol®). Attend to the product's application instruction with special regard to the concentration of the cleaner. Follow the directions in the EasyDens instruction manual.